Required files to uploaded
|Case Report|| |
Intra-operative fatigue failure of hexagonal screwdriver tip – case report of a potential complication in trauma surgery.
Ganesh Singh Dharmshaktu.
Intra-operative instrument breakage is a potential complication that not only leads to increased surgical time and agony to the patient but also is a source of frustration to the surgeon. The retrieval of the broken hardware may pose another challenge and sub optimal outcome in certain cases. There are numerous reports describing breakage of different sorts of medical devices and instruments and improvised methods of their successful retrieval. Orthopedic surgery requires use of manual force with instrumentation and thus the risk for important hardware failure remains omnipresent. The hexagonal screw-driver is widely used to fasten bone screws and is subjected to higher torque leading to rotation and torsion forces that may lead to excessive stress on it that at times goes past its yield point. The present case report describes an untoward event of breakage of screwdriver tip during surgery and its management. The presented report underlines importance of anticipation of this underreported complication of a common procedure.
Orthopedic Procedures, Complication, Trauma, Surgical Instruments, Equipment failure, Device removal, Fatigue failure.
|1. Lambotte A. Chirurgie Operatoire des Fractures. Paris, France: Masson Editors, 1913.|
|2. Pichler W, Mazzurana P, Clement H, Grechenig S, Mauschitz R, Grechenig W. Frequency of instrument breakage during orthopaedic procedures and its effects on patients. The Journal of bone and joint surg Am 2008; 90(12):2652-4. [DOI via Crossref] [Pubmed] |
|3. Price MV, Molloy S, Solan MC, Sutton A, Ricketts DM. The rate of instrument breakage during orthopaedic procedures. International orthopaedics 2002;26(3):185-7. [DOI via Crossref] [Pubmed] [PMC Free Fulltext] |
|4. Perren SM, Frigg R, Hehli M, et al. Lag screw. In: Ruedi TP, Murphy WM, eds. AO Principles of Fracture Management. Stuttgart: Thieme, 00.|
|5. Perren SM. Force measurements in screw fixation. J Biomech 1976;9:669-675. [DOI via Crossref] |
|6. Schatzker J, Sanderson R, Murnaghan JP. The holding power of orthopedic screws in vivo. Clin Orthop Relat Res 1975;108:115-126. [DOI via Crossref] [Pubmed] |